- Original research
- Open Access
A method for model-free partial volume correction in oncological PET
© Hofheinz et al.; licensee Springer. 2012
- Received: 23 January 2012
- Accepted: 24 April 2012
- Published: 24 April 2012
As is well known, limited spatial resolution leads to partial volume effects (PVE) and consequently to limited signal recovery. Determination of the mean activity concentration of a target structure is thus compromised even at target sizes much larger than the reconstructed spatial resolution. This leads to serious size-dependent underestimates of true signal intensity in hot spot imaging. For quantitative PET in general and in the context of therapy assessment in particular it is, therefore, mandatory to perform an adequate partial volume correction (PVC). The goal of our work was to develop and to validate a model-free PVC algorithm for hot spot imaging.
The algorithm proceeds in two automated steps. Step 1: estimation of the actual object boundary with a threshold based method and determination of the total activity A measured within the enclosed volume V. Step 2: determination of the activity fraction B, which is measured outside the object due to the partial volume effect (spill-out). The PVE corrected mean value is then given by C mean = (A+B)/V. For validation simulated tumours were used which were derived from real patient data (liver metastases of a colorectal carcinoma and head and neck cancer, respectively). The simulated tumours have characteristics (regarding tumour shape, contrast, noise, etc.) which are very similar to those of the underlying patient data, but the boundaries and tracer accumulation are exactly known. The PVE corrected mean values of 37 simulated tumours were determined and compared with the true mean values.
For the investigated simulated data the proposed approach yields PVE corrected mean values which agree very well with the true values (mean deviation (± s.d.): (−0.8±2.5)%).
The described method enables accurate quantitative partial volume correction in oncological hot spot imaging.
- Partial volume effect
- Partial volume correction
- Recovery correction
In recent years PET has become more and more important for therapy response assessment in oncology. In this context quantitation has been mostly restricted to assessment of changes of the maximum standardised uptake value (SUVmax) of lesions during therapy , but there are also attempts to correlate the SUVmean of lesions with therapy outcome, which might be a more representative parameter especially for lesions with heterogeneous tracer accumulation (see e.g. [2, 3]). However, the limited spatial resolution of PET leads to partial volume effects PVE and, consequently, to limited signal recovery for, both, SUV max and SUV mean . While SUVmax is affected only for small structures (whose size is comparable to – or smaller than – the given spatial resolution), SUVmean is compromised even at target sizes much larger than the reconstructed spatial resolution [4, 5]. Therefore, it is mandatory to perform an adequate PVE correction.
There exist several strategies for PVE correction (see[6–8] for recent reviews). Most often the PVE correction is computed on the basis of phantom measurements, where the signal recovery is determined for different object sizes and different background values (see e.g. [9–13]). The PVE correction is then performed using the signal recovery of a phantom with approximately the same volume and background as the target structure. Another approach is to improve spatial resolution either via deconvolution of the reconstructed PET data [14–19] or via integrating partial volume correction into the image reconstruction [18, 20–25]. A different strategy is to use model-free correction schemes, which directly determine the spill-out from the target structure but require knowledge of the object’s boundary and its background [26–28]. However, although many approaches have been shown to work in principle, there exists till now no general consensus regarding the best algorithm to use. Moreover, most algorithms are not generally available, neither in the public domain nor in commercial tools.
In this paper we present a model-free method for PVE correction of the SUVmean of focal structures. Our method can be considered as an extension of the methods reported in [26, 28]. In these papers, the object boundaries are determined in CT data and the mean value of a separate background ROI is considered as representative of the actual background of the target structure. Our extension is twofold: first, the object boundaries are determined directly in the PET data and second, for each voxel in the spill-out region a local background is computed independently instead of using a common background value for the complete ROI. For validation of the proposed approach the method was applied to simulated lesions, which were generated from (and embedded in) actual clinical patient data sets. The resulting “anthropomorphic digital phantoms” provide much more realistic conditions than conventional phantom measurements (which typically use regular shapes and homogeneous tracer distributions in target and background) and are visually not distinguishable from actual patient data. In the absence of a real gold standard we regard this as the best approach to evaluation of our algorithm.
Partial volume effect
The boundaries of the ROI are determined using the automatic ROI delineation method implemented in ROVER, ABX, Radeberg, Germany, which uses adaptive thresholding for ROI delineation (see  for details). The delineation also provides C mean and V ROI.
After ROI delineation the spill-out region is identified and for each voxel inside the spill-out region the local background is computed (see below). Then A sp is calculated according to Eq. (1).
Inserting the results from 1. and 2. in Eq. (2) leads to the PVE corrected mean value. In the following we refer to this algorithm as local background partial volume correction (LBPVC ). For comparison, we also compute the PVE corrected mean value using a global background for each ROI (see below). This algorithm is referred to as global background partial volume correction (GBPVC ).
The algorithm operates completely in 3D and runs fully automatic after ROI delineation. The computation time is less than one second (including volume delineation) per ROI on an AMD Opteron Processor (model 8356, 2.3 GHz).
The global background method uses a common background value for each ROI. The background region is defined in the same way as for the local background method. The global background value is then computed as the average value of the entire background region. Neighbouring ROIs as well as their spill-out regions are excluded from background determination. Note that this global background is different from the global background used in [26–28]: it is always computed in a matching background shell around the respective ROI even for irregular shaped ROIs and should, thus, provide a more realistic estimate of the actual average background of the given ROI.
The method was validated on simulated data. In such data the target structures have well known boundaries and tracer accumulation. The simulated data where created by modifications to a number of clinical PET data sets.
The simulated data before smoothing serve as our gold standard for which the true object boundaries, volumes, and SUVmean values are precisely known. In the following, we refer to these values as the true values. The Gaussian filter, applied to these data, then corresponds to an isotropic Gaussian point spread function with FWHM = 8 mm, which leads to approximately the same spatial resolution as in the original image data and, therefore, causes partial volume effects which are very similar to what is happening in real patient data.
Performance of the simulation procedure is illustrated in Figure 5. The original data are shown on the left. In the middle the resulting simulated artificial tumour with sharp boundary is shown, which serves as “ground truth” during evaluation of the algorithm. On the right the same structure after smoothing is shown which represents the “imaged” tumour, for which the PVC is to be evaluated. Figure 4 shows line profiles through the tumour along the grey lines indicated in Figure 5. The described simulation procedure leads to target structures which are very similar to their original counterparts with regard to several parameters such as mean and maximum uptake, target/background contrast, background characteristics, and the degree of heterogeneity (estimated as standard deviation of the mean value).
Simulation of low contrast structures and variable noise
In the chosen patient group, the simulated lesions obtained with the procedure described above exhibit contrasts (defined as ratio of maximum value to mean background) between 4.0 and 13.1, reflecting the actually observed conditions in this patient group. In order to study the influence of lower contrasts on LBPVC we reduced the voxel values inside the lesion by a factor of two for a subgroup of 5 selected lesions (while keeping surrounding voxels unmodified). In order to avoid secondary problems related to potential merging of lesion and nearby hot spots during the ensuing smoothing step only lesions without further hot spots in the immediate vicinity were selected for this procedure. The resulting additional 5 simulated lesions exhibited contrasts between 2.7 and 3.4. They were further analysed together with the initially simulated lesions in order to determine the contrast dependency of the partial volume correction procedure.We also performed the lesion simulation at three additional noise levels with a FWHM of the Gaussian noise of 3.5%, 7.1%, and 10%, respectively, augmenting the results obtained for FWHM = 5%. The additional noise levels were investigated in order to assess the noise sensitivity of the algorithm. This is of practical relevance, e.g., if scan times and injected doses are modified: in our case, decreasing the noise amplitude by a factor of to FWHM = 3.5% is equivalent to a doubling of the scan duration (or injected dose). Accordingly, increasing the noise amplitude by a factor of (2) to FWHM = 7.1% (10%) corresponds to reduction of the scan duration by a factor of 2 (4).
For the simulated data the PVE corrected SUVmean of the ROIs were determined using the LBPVC and GBPVC algorithms, respectively. The values were first computed using the known object boundaries (i.e., omitting the volume delineation step) and second by applying the complete correction scheme including volume delineation. In both cases, the corrected SUVmean was compared with the true SUVmean. Moreover, the automatically delineated ROI volumes were compared with the true volumes. For definition of the background and spill-out regions, we used a resolution value of FWHM = 8 mm. In order to test stability of the algorithm against uncertainties of the assumed resolution, we also performed evaluations with spill-out and background regions resulting from assuming resolution values of FWHM = 4 mm and FWHM = 12 mm, respectively. LBPVC and GBPVC correction was performed with the software ROVER (ABX GmbH, Radeberg, Germany).
Recovery correction using the true object boundaries
Recovery correction with automatic ROI delineation
Contrast and noise level dependency
Variation of the assumed spatial resolution
In this paper we present a model-free method for PVE correction of hot focal structures in PET. We have validated this method using realistic software phantoms of lesions generated from clinical data. The simulated lesions exhibit properties very similar to those of the underlying clinical data sets with respect to relevant parameters (shape/size, contrast, noise, etc.), while having precisely known boundaries and tracer accumulation. The simulated data allowed a direct comparison of the PVE corrected SUVmean resulting from automatic ROI delineation and application of LBPVC with the true SUVmean of the respective simulated lesions. We observe only a small difference between PVE corrected and true values (well below 10% in all cases, mean ± s.d.: (−0.8±2.5) %). This high accuracy is achieved by computing a local background for each voxel within a spill-out shell around the automatically delineated ROI. As is obvious from Figure 8, this approach (LBPVC) is superior to application of a common background value (GBPVC) especially for smaller objects with volumes below about 20 mL, even if the common background is determined in a matched background shell around respective ROI (as has be done here). The algorithm depends on a reasonable definition of a spill-out shell which contains all voxels whose activity values are elevated due to spill-out from the ROI plus a background shell whose voxels are assumed to be free of any spill-over effects. We have defined these shells in relative units using the estimated spatial resolution FWHM as the relevant length unit. Therefore, the assumed resolution does have a certain influence on the accuracy of the correction as demonstrated in Figure 11. If the estimate of FWHM is reduced by a factor of two, the necessary PVE correction is underestimated. This is explained by the fact that the spill-out region becomes too small and not all actually affected voxels are included. In this case, the procedure is therefore not able to collect the complete spill-out signal and, consequently, the correction is too small (especially for small structures).
On the other hand, an increase of the FWHM estimate by a factor of 1.5 results only in a slight overcorrection of the actual partial volume effect and the results are quite similar to those obtained with a realistic FWHM value (deviation from true value (4.3±3.8) %). Only for 4 out of 37 ROIs the deviation was larger than 10 % (but remained below 15 %) if the too large FWHM was adopted. It can thus be stated that the presented method leads to accurate results as long as the actual FWHM is not substantially underestimated. Accurate knowledge of the spatial resolution is not necessary, however. This is in contrast to e.g. deconvolution techniques [14–19], were the estimated resolution strongly influences the PVE correction. When in doubt (no accurate knowledge of actual spatial resolution), the best strategy, therefore, is to use a pessimistic (i.e. probably too high) estimate for FWHM, e.g. 8 mm even if actual resolution might be 6 mm.
Our method is similar to the methods discussed in [26, 28] with the important difference that our background approximation is local. This means the contribution of each voxel in the spill-out region to the PVE correction is computed using the background only in its immediate vicinity (up to a distance of 1.5· FWHM). In this way the method overcomes a limitation of the above mentioned methods which assume a homogeneous background for the whole ROI. In our approach we account for spatial variations of background intensity by determining an individual background level for each voxel and only assume that the background is homogeneous in the very small background area assigned to the respective voxel (blue area in Figure 2). This is a much weaker and more realistic assumption for most clinical PET studies. The superiority of LBPVC over GBPVC can be seen by comparison of Figure 8 (A) and (B). GBPVC (A) leads to reasonable results only for large ROIs (>20 mL). However, the corrected SUVmean of some of the small ROIs substantially deviate from the true values and ROI-to-ROI fluctuation is much higher than with LBPVC .
The proposed PVE correction critically depends on a sufficiently accurate estimate of the true object boundaries (without such an estimate, specification of SUVmean would not make sense anyway). For this task we used a threshold based automatic ROI delineation (see ). With this method we achieved good estimations of the true volume (deviation <10 %). The observed small deviations of the PVE corrected SUVmean from the true values are essentially an effect of the residual errors in the volume determination (or, rather, boundary delineation) alone. This is demonstrated by using the true object boundaries instead which is possible with our simulated target structures. In this case the difference between corrected and true values is nearly zero (see Figure 6B) which proves that the algorithm is able to correctly estimate the spill-over contributions from all voxels if the true boundary is known.
This shows, that accuracy of the presented PVE correction is essentially limited only by limitations of the used volume delineation process. Difficulties can, therefore, be expected especially for very small objects with diameter <2 · FWHM [4, 5]. In this case correct delineation and, therefore, reliable PVE correction method will certainly fail. A second limitation is the degree of heterogeneity in tracer uptake. Heavily heterogeneous ROIs cannot be delineated correctly with threshold based algorithms . The small but systematic underestimation of the delineated ROI volumes shown in Figure 7 can be attributed to this effect. In the present study the heterogeneities of the lesions were moderate (coefficient of intensity variance of voxels within the lesions: 0.13 to 0.22) and the errors in ROI delineation were very small, but it is clear that beyond a certain degree of heterogeneity the method will fail (although such problems can be expected in only a small percentage of the practically relevant cases). Further investigations are necessary to investigate the influence of larger heterogeneities in more detail.
Another factor principally limiting the accuracy of the volume delineation (and of the partial volume correction as well) is a too low contrast of the lesion. However, in our data, covering a contrast range from 2.7 to 13.1, we did not see a clear contrast dependency of the LBPVC correction as demonstrated in Figure 10. Nevertheless, we know from our experience in other investigations that the used delineation algorithm rapidly becomes unstable if the contrast falls below 2.5. Therefore, the presented correction method will not work reliable for such lesions.
A further factor influencing the accuracy of LBPVC is the noise level of the image data as demonstrated in Figure 9. This noise dependency is mainly a consequence of decreased accuracy of the volume delineation at elevated noise levels. Still, we found that the accuracy remains acceptable even if the noise level is doubled (corresponding to a fourfold decrease of scan time in comparison to our standard acquisition protocol): in this case only in 3 out of 37 lesions the error exceeds 10% (while remaining below 15%). It is obvious, however, that in the presence of excessive noise (e.g. in single gates from respiration triggered investigations) the presented correction method would not work reliably. Since the proposed correction algorithm does not require application of the specific delineation method used in this investigation, it could also be combined with alternative delineation algorithms with possibly improved performance (notably for heterogeneous structures). The correction method could of course also make use of available morphological information from CT or MRI for very small lesions or lesions with very low contrast, if available.
All investigated lesions with volumes in the range of 3 to 500 mL exhibited substantially reduced mean (as opposed to maximum) signal recovery. That this is the case even for large lesions is explained by the fact that the partial volume effect is a surface effect and the necessary PVE correction (regarding SUVmean) remains sizable even for rather large target structures. The partial volume effect is further increased for irregular/convoluted shapes (compared to approximately spherical objects of the same volume). Irregular shapes are of course not restricted to large structures. In our study sample most lesions with volumes > 10 mL were of distinctly irregular shape (see Figure 3). We consider the ability to perform accurate PVE correction for such structures as the most important benefit of the presented algorithm.
For the validation of our method we used simulated target structures which were derived from clinical data. The simulated target structures are much closer to real clinical data than typical (hardware or digital) phantoms. Realistic regional heterogeneities in the target structure or the background are especially difficult to realise (if at all) with the usual phantoms. The same is true regarding the generation/investigation of irregular shapes. Moreover, the standard spherical phantom inserts are hollow glass spheres whose cold walls can have a strong influence on the measured partial volume effects [31, 32] which are, therefore, not representative for the conditions found in real data. All these problems are avoided by our simulation procedure. For example, although the original tumour uptake heterogeneities are indeed modified by the smoothing applied during the simulation procedure (see above), they remain on a realistic level (see profiles in Figure 4). Despite absence of a true gold standard we believe that the performed validation allows to conclude that the proposed algorithm does provide a means for a quite accurate partial volume correction of real patient data.
The accuracy of the partial volume correction achieved in this study is comparable to the results reported in , where an accuracy better than 10% was found for lesions larger than 4 mL. The authors used simulated data and phantom measurements as ground truth and compared three different correction methods. However, all three investigated methods require a precise knowledge of the true point spread function (PSF) of the tomograph, while in our approach only a rough estimate of the PSF is needed. In  accurate correction capability is reported even for very small lesions (diameter 8 mm), but this method, too, requires a precise knowledge of the scanner’s PSF. We believe that requiring accurate knowledge of the PSF as a prerequisite is problematic and a potential source of substantial error of the partial volume correction, especially in a clinical context, where data sets might undergo individually different postprocessing/smoothing. Our approach, on the other hand is insensitive to variation of the actual PSF within a reasonable range of uncertainty which is an obvious advantage.Gallivan et al.  report on very good results without knowledge of the PSF, but only approximately spherical object with homogeneous tracer uptake were considered, which does not apply to the mostly irregular lesions observed in real patient data. Other authors have proposed to use anatomical information from high resolution CT or MRI (see e.g. [33–35]). This, however, requires very accurate coregistration of PET and CT/MRI which can be problematic even for modern PET/CT or PET/MRI systems due to patient motion during measurement. Probably more important, this approach rests on the assumption that the morphologically delineated lesion is identical to the hypermetabolic region observed in PET. As is well known, this assumption is by no means always correct. Such a lack of spatial concordance between morphological and functional signal would in turn lead to uncontrollable errors of the PVE correction. In this respect correction procedures relying exclusively on analysis of the PET data alone seem preferable.We, therefore, believe that the proposed method represents a viable, partly superior, alternative to other methods already discussed in the literature.
The presented approach to partial volume correction using local background determination distinctly improves quantitative accuracy of the correction in comparison to similar, previously described model-free approaches relying on a homogeneous background for the whole lesion. The improvement is especially pronounced for small lesions where the correction becomes numerically large. We conclude that adequate consideration of background heterogeneities on a per-voxel basis is mandatory to achieve reliable partial volume correction.
- Wahl R, Jacene H, Kasamon Y, Lodge M, et al.: From RECIST to PERCIST: Evolving Considerations for PET response criteria in solid tumors. J Nucl Med 2009,50(Suppl 1):122S-150S.PubMed CentralPubMedView ArticleGoogle Scholar
- Larson S, Erdi Y, Akhurst T, Mazumdar M, Macapinlac H, Finn R, Casilla C, Fazzari M, Srivastava N, Yeung H, et al.: Tumor Treatment Response Based on Visual and Quantitative Changes in Global Tumor Glycolysis Using PET-FDG Imaging: The Visual Response Score and the Change in Total Lesion Glycolysis. Clin Positron Imaging 1999,2(3):159–171. 10.1016/S1095-0397(99)00016-3PubMedView ArticleGoogle Scholar
- Denecke T, Rau B, Hoffmann K, Hildebrandt B, Ruf J, Gutberlet M, Huünerbein M, Felix R, Wust P, Amthauer H: Comparison of CT, MRI and FDG-PET in response prediction of patients with locally advanced rectal cancer after multimodal preoperative therapy: Is there a benefit in using functional imaging? Eur Radiol 2005,15(8):1658–1666. 10.1007/s00330-005-2658-4PubMedView ArticleGoogle Scholar
- Hoffman E, Huang S, Phelps M: Quantification in positron emission computed tomography. 1. Effect of object size. J Comp Assist Tomogr 1979,3(3):299–308. 10.1097/00004728-197906000-00001View ArticleGoogle Scholar
- Kessler R, Ellis J, Eden M: Analysis of emission tomographic scan data: limitations imposed by resolution and background. J Comp Assist Tomogr 1984,8(3):514–522. 10.1097/00004728-198406000-00028View ArticleGoogle Scholar
- Soret M, Bacharach S, Buvat I: Partial-volume effect in PET tumor imaging. J Nucl Med 2007,48(6):932. 10.2967/jnumed.106.035774PubMedView ArticleGoogle Scholar
- Rousset O, Rahmim A, Alavi A, Zaidi H: Partial volume correction strategies in PET. PET Clin 2007,2(2):235–249. 10.1016/j.cpet.2007.10.005View ArticleGoogle Scholar
- Gallivanone F, Stefano A, Grosso E, Canevari C, Gianolli L, Messa C, Gilardi M, Castiglioni I: PVE Correction in PET-CT Whole-Body Oncological Studies From PVE-Affected Images. Nucl Sci, IEEE Trans on 2011,58(3):736.View ArticleGoogle Scholar
- Avril N, Bense S, Ziegler S, Dose J, Weber W, Laubenbacher C, Römer W, Jänicke F, Schwaiger M: Breast imaging with fluorine-18-FDG PET: quantitative image analysis. J Nucl Med 1997,38(8):1186–1191.PubMedGoogle Scholar
- Präuer H, Weber W, Römer W, Treumann T, Ziegler S, Schwaiger M: Controlled prospective study of positron emission tomography using the glucose analogue [18f] fluorodeoxyglucose in the evaluation of pulmonary nodules. Br J Surgery 1998,85(11):1506–1511. 10.1046/j.1365-2168.1998.00915.xView ArticleGoogle Scholar
- Geworski L, Knoop B, de Cabrejas, Knapp W, Munz D: Recovery correction for quantitation in emission tomography: a feasibility study. Eur J Nucl Med Mol Imaging 2000,27(2):161–169. 10.1007/s002590050022View ArticleGoogle Scholar
- Aliaga A, Rousseau J, Cadorette J, Croteau É, van Lier J, Lecomte R, Bénard F: A small animal positron emission tomography study of the effect of chemotherapy and hormonal therapy on the uptake of 2-deoxy-2-[F-18] fluoro-d-glucose in murine models of breast cancer. Mol Imaging Biol 2007,9(3):144–150. 10.1007/s11307-007-0091-6PubMedView ArticleGoogle Scholar
- Sakaguchi Y, Mizoguchi N, Mitsumoto T, Mitsumoto K, Himuro K, Ohya N, Kaneko K, Baba S, Abe K, Onizuka Y, et al.: A simple table lookup method for PET/CT partial volume correction using a point-spread function in diagnosing lymph node metastasis. Anna Nucl Med 2010, 1–7.Google Scholar
- El Naqa, Low D, Bradley J, Vicic M, Deasy J: Deblurring of breathing motion artifacts in thoracic PET images by deconvolution methods. Med Phys 2006, 33: 3587. 10.1118/1.2336500View ArticleGoogle Scholar
- Teo B, Seo Y, Bacharach S, Carrasquillo J, Libutti S, Shukla H, Hasegawa B, Hawkins R, Franc B: Partial-volume correction in, PET validation of an iterative postreconstruction method with phantom and patient data. J Nucl Med 2007,48(5):802.PubMedGoogle Scholar
- Kirov A, Piao J, Schmidtlein C: Partial volume effect correction in PET using regularized iterative deconvolution with variance control based on local topology. Phys Med Biol 2577, 53: 2008.Google Scholar
- Barbee D, Flynn R, Holden J, Nickles R, Jeraj R: A method for partial volume correction of PET-imaged tumor heterogeneity using expectation maximization with a spatially varying point spread function. Phys Med Biol 2010, 55: 221. 10.1088/0031-9155/55/1/013PubMed CentralPubMedView ArticleGoogle Scholar
- Hoetjes N, van Velden, Hoekstra O, Hoekstra C, Krak N, Lammertsma A, Boellaard R: Partial volume correction strategies for quantitative FDG PET in oncology. Eur J Nucl Med Mol Imaging 2010,37(9):1679–1687. 10.1007/s00259-010-1472-7PubMed CentralPubMedView ArticleGoogle Scholar
- van Velden F, Cheebsumon P, Yaqub M, Smit E, Hoekstra O, Lammertsma A, Boellaard R: Evaluation of a cumulative SUV-volume histogram method for parameterizing heterogeneous intratumoural FDG uptake in non-small cell lung cancer PET studies. Eur J Nucl Med Mol Imaging 2011, 38: 1636. 10.1007/s00259-011-1845-6PubMed CentralPubMedView ArticleGoogle Scholar
- Brix G, Doll J, Bellemann M, Trojan H, Haberkorn U, Schmidlin P, Ostertag H: Use of scanner characteristics in iterative image reconstruction for high-resolution positron emission tomography studies of small animals. Eur J Nucl Med Mol Imaging 1997,24(7):779–786.Google Scholar
- Reader A, Julyan P, Williams H, Hastings D, Zweit J: EM algorithm system modeling by image-space techniques for PET reconstruction. Nucl Sci, IEEE Trans on 2003,50(5):1392–1397. 10.1109/TNS.2003.817327View ArticleGoogle Scholar
- Alessio A, Kinahan P, Lewellen T: Modeling and incorporation of system response functions in 3-D whole body PET. Med Imaging, IEEE Trans on 2006,25(7):828–837.View ArticleGoogle Scholar
- Vandenberghe S, Karp J: Rebinning and reconstruction techniques for 3D TOF-PET. Nucl Instrum Methods Phys Res A 2006,569(2):421–424. 10.1016/j.nima.2006.08.065View ArticleGoogle Scholar
- Rizzo G, Castiglioni I, Russo G, Tana M, Dell Acqua, Gilardi M, Fazio F, Cerutti S: Using deconvolution to improve PET spatial resolution in OSEM iterative reconstruction. Methods Inf Med 2007,46(2):231.PubMedGoogle Scholar
- Tohme M, Qi J: Iterative image reconstruction for positron emission tomography based on a detector response function estimated from point source measurements. Phys Med Biol 2009, 54: 3709. 10.1088/0031-9155/54/12/007PubMed CentralPubMedView ArticleGoogle Scholar
- Hickeson M, Yun M, Matthies A, Zhuang H, Adam L, Lacorte L, Alavi A: Use of a corrected standardized uptake value based on the lesion size on CT permits accurate characterization of lung nodules on FDG-PET. Eur J of Nucl Med Mol Imaging 2002,29(12):1639–1647. 10.1007/s00259-002-0924-0View ArticleGoogle Scholar
- Verel I, Visser G, Boellaard R, Boerman O, van Eerd, Snow G, Lammertsma A, van Dongen: Quantitative 89Zr immuno-PET for in vivo scouting of 90Y-labeled monoclonal antibodies in xenograft-bearing nude mice. J Nucl Med 2003,44(10):1663.PubMedGoogle Scholar
- Bundschuh R, Essler M, Dinges J, Berchtenbreiter C, Mariss J, Martínez-Möller A, Delso G, Nekolla S, Schulz D, Hohberg M, Hohberg M, et al.: Semiautomatic Algorithm for Lymph Node Analysis Corrected for Partial Volume Effects in Combined Positron Emission Tomography-computed Tomography. Mol Imaging 2010,9(6):319.PubMedGoogle Scholar
- Hofheinz F, Pötzsch C, Oehme L, Beuthien-Baumann B, Steinbach J, Kotzerke J, van den Hoff J: Automatic volume delineation in oncological PET. Evaluation of a dedicated software tool and comparison with manual delineation in clinical data sets. Nuklearmedizin Nucl Med 2011., 51: Google Scholar
- Hatt M, Lamare F, Boussion N, Turzo A, Collet C, Salzenstein F, Roux C, Jarritt P, Carson K, Cheze-Le Rest, Visvikis D: Fuzzy hidden Markov chains segmentation for volume determination and quantitation in PET. Phys Med Biol 2007, 52: 3467. 10.1088/0031-9155/52/12/010PubMed CentralPubMedView ArticleGoogle Scholar
- Bazanez-Borgert M, Bundschuh R, Herz M, Martinez M, Schwaiger M: Radioactive spheres without inactive wall for lesion simulation in PET. Zeitschrift für medizinische Physik 2008, 18: 37–42. 10.1016/j.zemedi.2007.06.001PubMedView ArticleGoogle Scholar
- Hofheinz F, Dittrich S, Pötzsch C, van den Hoff J: Effects of cold sphere walls in PET phantom measurements on the volume reproducing threshold. Phys Med Biol 2010,55(4):1099–113. 10.1088/0031-9155/55/4/013PubMedView ArticleGoogle Scholar
- Rousset O, Ma Y, Evans A, et al.: Correction for partial volume effects in PET: principle and validation. J Nucl Med: Official Publ, Soc Nucl Med 1998,39(5):904.Google Scholar
- Alessio A, Kinahan P: Improved quantitation for PET/CT image reconstruction with system modeling and anatomical priors. Med Phys 2006, 33: 4095. 10.1118/1.2358198PubMedView ArticleGoogle Scholar
- Boussion N, Hatt M, Lamare F, Bizais Y, Turzo A, Cheze-Le Rest C, Visvikis D: A multiresolution image based approach for correction of partial volume effects in emission tomography. Phys Med Biol 2006, 51: 1857. 10.1088/0031-9155/51/7/016PubMedView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.